Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
|Published (Last):||11 March 2013|
|PDF File Size:||19.53 Mb|
|ePub File Size:||7.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
October Softclix are trademarks of a Member of the Roche Group. It is best to have your instrument next to you while reading the manual and to try out each step as it is described.
If any of these items are missing or damaged please contact your supplier. If at any time during the function test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and The time must also be changed from summer to win- ter time and vice versa, and 29 February must be entered in a leap year.
The hours flash on the display. Press the SET button kanual. The minutes flash on the display. Keep pressing to set the minutes in the same way as the hour. If the instrument has not been coded before, the display shows.
When the test CODE strip is inserted the display automatically switches to the cor- rect symbol for the strip. Users in the office and laboratory sector should note that samples containing human material must be handled as potentially infectious material. Then CODE stops flashing. If the strip code has not been read in correctly, error messages appear e.
If too little blood is applied, do not try to spread it or apply a second drop as this avcutrend also produce incorrect results.
This can lead to inappropriate treatment rec- ommendations and thus endanger your health. If other messages appear on the display, consult Sec- Too little blood applied — The blood must be applied to the test strips outside the meter. After lancing the finger remove the test strip from the open gvt and place on a piece of clean, absorbent material, e.
The relevant regulations or guidelines applying to your institute or laboratory should be observed. The used test strips and consumables e. Safety warnings such as are also stored. If the date and time have been set Section 2. Select the relevant directory by pressing for glucose values or for cholesterol values.
Press the button MEM and hold it down to run through all memory positions.
Roche Accutrend GC User Manual
When all positions have been shown the display returns to To count down from any value stored: You have the following possibilities: If the button is released earlier the clearing process will be interrupted. A performance check should be carried out: The procedure for checking the measuring system is identical to the procedure for practising how to perform a measure- ment.
Make sure that the strip is inserted as far as it will go and that the word GLU on the strip can no longer be seen. CODE The display now counts down in seconds to. During the last four seconds a series of beeps is heard one beep per second followed by a longer beep at the end of the measurement.
If the result is again accutrens the specified range, consult your nearest Ch. It should also be cleaned: They are essential to ensure precision and reliability of the measurements. Humidity In the presence of high humidity or large temperature changes condensation may form mqnual the meter, e. Move the meter to maunal dry environment and wait a while before switching it on again. Display remains empty when you switch on the instrument The meter is defective. Contact the Roche Diagnostics service centre.
Code the meter see Section 3. Have manial batteries ready.
Always change all three batteries at the same time see Section 5. Recode the meter with the code strip from the pack of test strips you are using see Section 3. The test strip is not suitable for the meter. Move the meter to a different place see Section 5. accutrenv
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
The optics are dirty or the meter is defective. Clean the meter see Section 5. If the problem persists, contact the Roche Diagnostics service centre see Section 6.
The round window on the back of the strip is not uniformly coloured. Use accytrend fresh test strip. The test strip is too old or has been incorrectly stored. Check the expiry date and storage conditions of the test strips. Ready for setting the date and time or the beeper. If you suspect a fault in your meter you should first contact the advisory service at your nearest Roche Diagnostics office.
Additional information for medical personnel and laboratories 7. Light from an LED light emitting diode is accturend towards the test strip from below and the blank value of the test strip determined by measurement of the reflected light. For further information on this Weight: Liquid crystal display Automatic cut-off: After 2 minutes with test strip inserted; after 1 minute if no button was pressed and no test strip has been inserted.
For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office. Index Page Roche Diagnostics K.
Blood collectionTiong Bahru Road Plausibility check 37, 69 Practising measurement 29ff, 59 Retrieval of results from memory 49ff. Storage conditions Switching on Symbols 4, 5 Technical data Test procedure 29ff. Page of 58 Go. Illustrations of the meter are shown on the fold-out page.
The abbreviations used by the Accutrend. Table of Contents Add to my manuals Add. Page 8 If at any time during the function test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and Page 36 If the result is again outside the specified range, consult your nearest Ch.
Troubleshooting Move the meter maanual a dry environment and wait a while before switching it on again. Page 46 Move the meter to a different place see Section 5.
Software x 62 x System Components For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office. Manufacturer’s Warranty And Guarantee Comments to this Manuals Your Name.
Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. Don’t show me this message again. Got it, continue to print. Don’t have an account?